Main Article Content
This paper presents a new design of low cost 12-lead Electrocardiogram (ECG) acquisition system. The system is an integration of a biosignal analog-to-digital front-end
evaluation board, supplied by Texas InstrumentTM, and a widely known controller from Raspberri PITM. These two separately available gadgets were connected and operated by the costom-made software. This acquisition system is made for long-term multi-channel physiological signals recording with its own storage. With no need of an extra computer for operating, the system is stand-alone working at low cost. For more than two hours, our prototype unit was running through a test, acquiring 12-lead ECG signal from a simulator at 500Hz sampling frequency. The restored signals were constantly confirmed. The target groups of this work are the educators and researchers in biomedical engineering area who are demanding the biopotential signals. Our future work is to develop a new user interface and data analysis software.
Article Accepting Policy
The editorial board of Thai-Nichi Institute of Technology is pleased to receive articles from lecturers and experts in the fields of business administration, languages, engineering and technology written in Thai or English. The academic work submitted for publication must not be published in any other publication before and must not be under consideration of other journal submissions. Therefore, those interested in participating in the dissemination of work and knowledge can submit their article to the editorial board for further submission to the screening committee to consider publishing in the journal. The articles that can be published include solely research articles. Interested persons can prepare their articles by reviewing recommendations for article authors.
Copyright infringement is solely the responsibility of the author(s) of the article. Articles that have been published must be screened and reviewed for quality from qualified experts approved by the editorial board.
The text that appears within each article published in this research journal is a personal opinion of each author, nothing related to Thai-Nichi Institute of Technology, and other faculty members in the institution in any way. Responsibilities and accuracy for the content of each article are owned by each author. If there is any mistake, each author will be responsible for his/her own article(s).
The editorial board reserves the right not to bring any content, views or comments of articles in the Journal of Thai-Nichi Institute of Technology to publish before receiving permission from the authorized author(s) in writing. The published work is the copyright of the Journal of Thai-Nichi Institute of Technology.
“Heart rate variability. Standards of measurement, physiological interpretation, and clinical use. Task Force of the European Society of Cardiology and the North American Society of Pacing and Electrophysiology,” Eur. Heart J., vol. 17, no. 3, pp. 354-381, Mar.1996.
A. M. Nasrabadi and M. H. Kani, “Design of ECG acquisition and transmission via Bluetooth with heart disease diagnosis,” in 2011 IEEE International Workshop on Medical Measurements and Applications Proceedings (MeMeA), 2011, pp. 55-58.
C. Sun, J. Liao, G. Wang, B. Li, and M. Q.-H. Meng, “Design of ECG acquisition and transmission via Bluetooth with heart disease diagnosis,” in IEEE International Conference on Information and Automation, 2013, p. 368.
W. Baochun, L. Min, Y. Yaning, and Z. Weiwei, “ECG acquisition circuit design based on C8051F330,” in Proceedings of 2012 IEEEEMBS International Conference on Biomedical and Health Informatics, 2012, pp. 93–96.
Z. Gao, J. Wu, J. Zhou, W. Jiang, and L. Feng, “Design of ECG Signal Acquisition and Processing System,” in Proceedings of the 2012 International Conference on Biomedical Engineering and Biotechnology, Washington, DC, USA, 2012, pp. 762-764.
J. A. G. Gnecchi, A. d J. V. Herrejón, A. d C. T. Anguiano, A. M. Patiño, and D. L. Espinoza, “Advances in the Construction of ECG Wearable Sensor Technology: The ECG-ITM-05 eHealth Data Acquisition System,” in Electronics, Robotics and Automotive Mechanics Conference (CERMA), 2012 IEEE Ninth, 2012, pp. 338-342.
G. D. Gargiulo, P. Bifulco, M. Cesarelli, A. Mcewan, and A. Wabnitz, “Open platform, 32-channel, portable, data-logger with 32 pga control lines for wearable medical device development,” Electronics Letters, vol. 50, no. 16, pp. 1127–1129, Jul. 2014.
A. J. Cook, G. D. Gargiulo, T. Lehmann, and T. J. Hamilton, “Open platform, eight-channel, portable bio-potential and activity data logger for wearable medical device development,” Electronics Letters, vol. 51, no. 21, pp. 1641–1643, 2015.
F. Abtahi, J. Snäll, B. Aslamy, S. Abtahi, F. Seoane, and K. Lindecrantz, “Biosignal PI, an Affordable Open-Source ECG and Respiration Measurement System,” Sensors, vol. 15, no. 1, pp. 93-109, Dec. 2014.
“Low-Power, 8-Channel, 24-Bit Analog Front-End for Biopotential Measurements,” 2015. [Online]. Available: https://www.ti.com.
ADS1298RECG-FE ECG Front-End Performance Demonstration Kit User’s Guide,” 2015. [Online]. Available: https://www.ti.com.
“Altium Limited,” 2015. [Online]. Available: https://www.altium.com.